I was wondering what everyone used as
I was wondering what everyone used as their calibration limits for humidity linked to their stability chambers?
I am using a ± 1.0%RH limit for the Relative Humidity Transmitter, the Relative Humidity Controller and the Chart Recorder Humidity, but am wondering whether this is too tight.
Thanks for your help!
Senior Manager, Stability & Quality Control Systems
24 Ivey Lane, PO Box 2880
Windsor, Nova Scotia
THE INFORMATION CONTAINED IN THIS COMMUNICATION AND ANY ATTACHMENTS
HERETO IS CONFIDENTIAL, MAY BE ATTORNEY-CLIENT PRIVILEGED, AND IS
INTENDED ONLY FOR THE PERSONAL AND CONFIDENTIAL USE OF THE
ADDRESSEE(S). IF THE READER OF THIS MESSAGE IS NOT AN INTENDED
RECIPIENT, OR AN AGENT THEREOF, YOU ARE HEREBY NOTIFIED THAT ANY
REVIEW, USE, DISSEMINATION, DISTRIBUTION, OR COPYING OF THIS
COMMUNICATION OR ANY ATTACHMENT HERETO IS STRICTLY PROHIBITED. IF
YOU HAVE RECEIVED THIS MESSAGE IN ERROR, PLEASE NOTIFY US
IMMEDIATELY BY E-MAIL, AND DELETE THE ORIGINAL MESSAGE.
The nature of this service is to provide a medium for communication. The specific statements and endorsements of individuals participating in the discussions are not necessarily those of The Microbiology Network, Inc., the PSDG, or the sponsors of the list.
The PSDGList (http://microbiol.org/psdglist.htm) is operated from The Microbiology Network (http://www.microbiol.org) and supported by
our sponsors as a service to the scientific community. Please take a second to visit our sponsors' web sites and say thank you for their support of this service.
Bahnson Environmental Specialties - http://www.eschambers.com/
Veriteq Instruments - http://www.veriteq.com/
Vectech Pharmaceutical Consultants, Inc - http://www.vectech.com